Replacement Electrode for Combo Meter P160 Pro
The P160 Pro uses a replaceable electrode. This allows for accurate measurement of pH and EC values in an aqueous solution. The electrode is designed to be easily replaced when it is worn out or its accuracy is compromised due to dirt or damage.
In the user manual, you will find instructions on how to replace the electrode. First, ensure that the combo meter is turned off. Now, carefully remove the electrode from the device. Then, you can insert the new electrode.
Another feature of the Combo Meter P160 Pro is that the electrode is waterproof according to IP67 standard. This means it is protected from water ingress and can be used in humid environments or underwater.
It's important to note that the lifespan of the electrode depends on the frequency of your measurements and the type of solutions it is used in. Therefore, it is recommended to regularly check, clean, and replace the electrode if necessary.
Instructions for Replacing the Electrode of the Combo Meter P160 Pro
The replacement can be easily done by yourself when the electrode is worn out or damaged.
- Remove the cover, loosen the sealing rings, and gently pull out the electrode.
- The electrode must be carefully reinserted, and the sealing ring must be securely fastened.
- The combo meter should be recalibrated after replacing the electrode and will be ready for use again.
Included in the Package
- 1x Replacement Electrode for Combo Meter P160 Pro